Weekdays 10:00 - 19:00 (closed on weekends and holidays)
We are open to adding new features based on your or your clients' needs to ensure the better experience.
Contact : firstname.lastname@example.org
00. The minimum space required for posture analysis is provided below, but it is recommended that the space for dynamic assessment be 1.5 times larger
To ensure accurate results, please make sure there is at least 50cm of space between the equipment and both side and back walls. Additionally, be sure not to block the camera's field of view. Having more space is advantageous for achieving accurate results.
01. Set your mat on a flat surface
If the floor where the camera stand placed is uneven, the measurable height can be reduced by up to 160 cm. Also note that uneven floors can cause errors.
02. Avoid exposure to sunlight when taking measurements
The camera of Moti-Physio uses infrared light. Sunlight contains a lot of infrared light, which can confuse the sensor part of the camera that reads infrared light. Therefore, please use in an environment that does not expose you to sunlight (such as using a blackout curtain). Additionally, please place it in a place where there is no reflective material such as a mirror.
03. Clear the area around the mat
Errors may occur if there are walls around where you are standing, so make sure you have enough space.
Set the minimum separation distance between the mat and the side or back wall where you are standing to 50cm.
[System requirements for Installing Moti-Physio Software]
OS: Windows 10 64bit or Windows 11 64bit (MacOS unavailable currently)
CPU: Intel® Core™ i5 10th ⬆︎ (“Not Available : AMD Radeon R7 M260” )
GPU: Intel® UHD Graphics 630 ⬆︎
USB Port: USB2.0 or USB3.0 1ea
RAM: 8GB ⬆︎
Storage: 10GB ⬆︎
Internet should be always connected
Resolution: 1920 x 1080 (fixed)
If you are interested in becoming a distributor or local sales partner, please contact us via email (email@example.com) and provide us with your company profile.
We have received many purchase inquiries from North America and the EU and we want to share the revenue in each market.
A. Static Analysis: Using an RGB-D camera, we capture front, side, back, and Adam's Test images, and analyze skeletal asymmetry through deep learning. Based on the analyzed data, we evaluate 11 items such as Forward Head Posture, Asymmetrical Shoulder Height, C7-CSL x-axis distance, and Pelvic Axial Rotation, using physiotherapy theories such as Janda's Approach and Anatomy Train. Then, we analyze the relative tension and weakness of 87 muscles. Finally, we create a digital twin of customers' body data and reconstruct the 3D skeleton and muscles.
B. Dynamic Analysis: Actions such as the Overhead Squat Assessment and Single Leg Stand required, and deep learning used to analyze the pattern of each joint's movement for results.
C. Additional features will be continuously updated in the future.
No, MOTI PHYSIO is not certified as a medical device yet.
Medical device certification has different definitions and procedures in each region. We plan to launch a Medical Device Certification for Moti-Physio in 2024.(CE medical, FDA)
Currently, Moti-Physio is not medical device. So, its purpose is only screening & monitoring for patients’ rehabilitation status or process of musculoskeletal care service.
Moti-Physio is not a medical device, but it is designed for screening and monitoring patients' rehabilitation status or musculoskeletal care services.
Moti Physio's result shows in degrees. As a result of the experiment, when the error test was performed with the mannequin standing up, the maximum error was about 1.5 degrees.
Clothing fits are important for Moti Physio. We recommend wearing tight clothing as much as possible. And you have to test in a space without sunlight to get an accurate result.
There is no human error in Moti Physio because of no intervention of the practitioner. And intra reliability is explained by a pearson correlation coefficient. Its value is 0.6-0.8 in each parameter. This clinical trial was conducted on 100 people. This result has been published in two SCIE-level journals.
< References >
A. Appraisal of the New Posture Analyzing and Virtual Reconstruction Device (PAViR) for Assessing Sagittal Posture Parameters: A Prospective Observational Study
B. Validity and Reliability of a Non-Radiographic Postural Analysis Device Based on an RGB-Depth Camera Comparing EOS 3D Imaging: A Prospective Observational Study
Korean, English, Japanese version are available currently as of March 2023
Spanish, Italian, French, German, Chinese will be available in September 2023
Yes for a product with defect by our problem.
You have 30 calendar days to return an item from the date you received it. Follow these steps:
A. Contact our customer support team via email or phone to request a refund.
B. Provide a valid reason for the refund request, along with any relevant documentation or evidence.
C. Return the product in its original packaging, along with all included accessories and materials, in the same condition as when you received it. You will be responsible for the return shipping costs first and will be refunded after confirmation of defect.
D. Once we receive and inspect the returned product, we will process the refund within 7 business days. The refund will be issued to the original payment method used during the purchase.
However, if the reason for the refund is not valid, you will be responsible for the shipping cost.
Exceptions to the refund policy may apply for certain products or services, which will be clearly stated at the time of purchase. We reserve the right to refuse a refund request if it does not comply with our refund policy.